ZPower's Battery Expert Answers the Most Commonly Asked Questions!
ZPower silver-zinc rechargeable batteries are a new technology now available for select products from almost every major hearing aid manufacturer. With new technology comes many questions, so we have introduced an “Ask the Expert” program to help you get the most out of your hearing aid’s silver-zinc rechargeable system.
How often should the batteries be charged?
The batteries should be fully charged every night. Once the hearing aids are finished charging, the indicator lights turn from blinking green to solid green. A full charge may take up to 7 hours — the charge time varies based on how much the battery was depleted during the day.
What happens when the silver-zinc rechargeable battery is getting low on power?
The hearing aid wearer will hear the low battery warning. Once the low battery warning occurs or once a hearing aid shuts off due to a low battery condition, do not open the battery door to reboot the hearing aid. Rebooting after the low battery warning can over-discharge the battery. If a low battery warning from the hearing aids is received, place the hearing aids in the charging base for charging or replace with non-rechargeable batteries. Store your rechargeable batteries in a safe place away from metal objects like coins or keys.
Should a patient open the battery door to turn off the hearing aids when not in use?
If a patient removes their hearing aids during the day, the hearing aids should be put back in the charger. Rule of thumb: If the hearing aids are not on the ears, they should be in the charger. The batteries will not overcharge, and this will not decrease the overall life of the battery. If the charger is not available, it’s ok to open the battery door for a few hours until they are worn or can be put in the charger. If the hearing aids are not going to be worn for an extended period, the batteries should be removed.
Can I use an active drying system for hearing aids with ZPower batteries?
Active drying systems may have temperature ranges that exceed the recommended rechargeable battery operating temperature, which can cause damage to silver-zinc rechargeable batteries. Therefore, when using an active drying system, the batteries need to be removed from the hearing aids prior to drying. For this reason, ZPower has partnered with MG to bring a compatible dryer system to the market. During the late summer of 2018, MG will be launching a retrofit drying system that will work with any ZPower charger. This system will provide charging, active drying and a UV light - all in one system.
Do I need to clean the charging contacts on my hearing aids and charger?
Yes, to ensure your hearing aids charge properly, it is important that the hearing aids and charger are dry and clean before charging. Use a soft tissue to remove moisture or debris on the hearing aids and charger contacts. The soft brush that is included with hearing aids may also be used to remove debris from the charging wells. Do not use rubbing alcohol or other chemical substances.
Watch Our Video – Getting the Most Out of Your ZPower Rechargeable System
Still Have Questions?
We're Here to Help - Fill Out Our Inquiry Form and a ZPower Expert Will Get Back to You Shortly.Ask the Expert